Measure total reduced sulfur (TRS) using field-proven, pulsed fluorescence technology with Thermo Scientific™ Continuous Emissions Monitoring Systems (CEMS).
Designed to meet the needs of the pulp and paper industry in complying with the regulatory requirements of 40 CFR part 60 as set forth by the U.S. EPA.
In order to measure TRS in accordance with the federal regulations, the sample must be measured on a dry basis and be corrected to a fixed Oxygen (O2) content. This is accomplished by extracting the sample from the process using the Thermo Scientific PRO902C Low-Flow Dilution Probe.
First, the sample is passed through a water removal device to provide a dry sample. The oxygen content is measured and the sample is diluted to an appropriate level, depending on the site conditions.
The diluted sample is sent through a sample line to the instrument cabinet and passes through a selective scrubber where Sulfur Dioxide (SO2) is removed and TRS is allowed to pass. Next, the sample is routed to a thermal oxidizer where the TRS compounds react with O2 under high temperature, and is then sent on to the Model 43i Pulsed Fluorescent SO2 analyzer.
Pulp and Paper Mills
Review your order status.
Overview of all services including support plan options, repairs and calibrations, spare parts and technical training.
Request an RMA number and send your instrument to our factory for repair.
Sign up to receive tips delivered to your inbox
Our technical and engineering staff share their product and application expertise in Fast5 Tips and Tricks for gas analyzers and systems. Fast5 offers a tip or trick that you can read in five minutes or less.