QuantStudio™ 12K Flex Real-Time PCR System – Troubleshooting
Having difficulties with your experiment?
We are dedicated to your success. Get back on track. View our expert recommendations for commonly encountered problem scenarios.
View the relevant questions below:
Sometimes an error can occur when the file is downloaded to the computer. When this occurs you may see an abrupt break in the data, like in the example below (Before image). If you see this, you can repair the data by going to Analysis → Analyze from Raw Data. You should then see a fixed plot like the ‘After’ image. Please note that you would need to have at least v1.2 of the QuantStudio™ 12K Flex software to use this feature. If you do not have this version, or if you do but the error persists, please send your data file to firstname.lastname@example.org so that we can attempt to recover the data.
This error can occur due to allocation of the computer’s virtual memory. If you see this message, try the following:
- Click ‘OK’ to the error, close any unnecessary programs, and relaunch the software (you may have to do this several times).
- Using the Task Manager (Control+Alt+Delete), look for a process called ‘javaw.exe’. Select this line and end the process. Then relaunch the software.
When this happens, the analysis was set to use ‘Relative Threshold’ but the CRT values do not show up in the exported data. (Note: To use ‘Relative Threshold’, go to “Analysis Settings”, and under the default Ct Settings tab, choose ‘Relative Threshold’ under “Algorithm Settings”.)
To export the data, go to ‘Export’ and make sure the ‘Results’ section is checked. You will notice that the column header will still say ‘Ct’ even if you have CRT in the well table like above. If you export the data but notice that the numbers still correspond to the ‘Ct’ values (and not the CRT values), then you will need to use the ‘Load Export Set’ button. Just click ‘OK’ to the pop-up window. and the changes will be applied. The next time you export, the values should reflect the CRT values.
Please follow the instructions found in this User Bulletin: How to Activate or Renew a License. (Please note that these instructions still apply, even though the example is using the ViiA™ 7 instrument.) If you have a license key associated with the computer that is no longer in use, please contact our Instrument Services group at email@example.com with the serial number and license key so that they can reassign it for you.
For Research Use Only. Not for use in diagnostic procedures.