The iQ Series of gas analyzers comes standard with Ethernet and USB communication capability. If you require a Serial, Analog or Digital connection, you will either need to order one or more of the optional I/O boards at the time of initial purchase or they can be easily installed post-initial instrument sale. After installation, make sure the newly installed board is enabled in the “Configuration” menu. Once the change is committed in the “Configuration” menu, you will now have access to the I/O menu for that board under “Communications.” Before you can take advantage of the I/O capability, you will need to set up each required channel via the “Communications” menu. Here is an example of how to set up an Analog voltage out channel.
Analog voltage out:
PLEASE NOTE: The min & max value is not a % value, it is a discrete range value entered in instrument base units, either ppb or µg/m3.
Example: Maximum concentration range is 20 ppm, the max limit value should be 20,000.
How to calibrate an analog voltage channel in an iQ instrument:
Visit Thermofisher.com/iqseries for each instrument’s individual product page. Scroll to the bottom of the page and under Documents, find the user manual. Download and view the Optional Equipment section, then see Connecting External Devices to review.
Schedule your analyzer, calibrator, probe controller or probe for a Return Material Authorization (RMA) to factory depot repair for complete maintenance and factory specification service.