Electrophoresis is a standard laboratory technique by which charged molecules are transported through a solvent by an electrical field. Both proteins and nucleic acids may be separated by electrophoresis, which is a simple, rapid, and sensitive analytical tool. Most biological molecules carry a net charge at any pH other than their isoelectric point and will migrate at a rate proportional to their charge density. The mobility of a molecule through an electric field will depend on the following factors: field strength, net charge on the molecule, size and shape of the molecule, ionic strength, and properties of the matrix through which the molecule migrates (e.g., viscosity, pore size). Polyacrylamide and agarose are two support matrices commonly used in electrophoresis. These matrices serve as porous media and behave like a molecular sieve. Agarose has a large pore size and is suitable for separating nucleic acids and protein complexes. Polyacrylamide has a smaller pore size and is ideal for separating most proteins and smaller nucleic acids.
What is protein electrophoresis?
Several forms of polyacrylamide gel electrophoresis (PAGE) exist, and each form can provide different types of information about proteins of interest. Denaturing and reducing sodium dodecyl sulfate (SDS)-PAGE with a discontinuous buffer system is the most widely used electrophoresis technique and separates proteins primarily by mass. Nondenaturing PAGE, also called native PAGE, separates proteins according to their mass/charge ratio. Two-dimensional (2D) PAGE separates proteins by native isoelectric point in the first dimension and by mass in the second dimension.
SDS-PAGE separates proteins primarily by mass because the ionic detergent SDS denatures and binds to proteins to make them uniformly negatively charged. Thus, when a current is applied, all SDS-bound proteins in a sample will migrate through the gel toward the positively charged electrode. Proteins with less mass travel more quickly through the gel than those with greater mass because of the sieving effect of the gel matrix.
Watch this video summary protein gel electrophoresis by SDS-PAGE.
Once separated by electrophoresis, proteins can be detected in a gel with various stains, transferred onto a membrane for detection by western blotting and/or excised and extracted for analysis by mass spectrometry. Protein gel electrophoresis is, therefore, a fundamental step in many kinds of proteomics analysis.
Protein Gel Electrophoresis Technical Handbook
This 88-page handbook provides detailed description about all aspects of protein electrophoresis from sample and gel preparation to choice of molecular weight markers. In addition, it contains extensive information about our portfolio of high-quality protein electrophoresis products including gels, stains, molecular weight markers, running buffers, and blotting products for your experiments.
Acrylamide is the material of choice for preparing electrophoretic gels to separate proteins by size. Acrylamide mixed with bisacrylamide forms a crosslinked polymer network when the polymerizing agent, ammonium persulfate (APS), is added. TEMED (N,N,N,N'-tetramethylenediamine) catalyzes the polymerization reaction by promoting the production of free radicals by APS.
Polymerization and crosslinking of acrylamide. The ratio of bisacrylamide (N,N'-methylenediacrylamide) to acrylamide, as well as the total concentration of both components, affects the pore size and rigidity of the final gel matrix. These, in turn, affect the range of protein sizes (molecular weights) that can be resolved.
Example recipe for a traditional 10% Tris-glycine mini gel for SDS-PAGE:
- 7.5 mL 40% acrylamide solution
- 3.9 mL 1% bisacrylamide solution
- 7.5 mL 1.5 M Tris-HCl, pH 8.7
- Add water to 30 mL
- 0.3 mL 10% APS
- 0.3 mL 10% SDS
- 0.03 mL TEMED
The size of the pores created in the gel is inversely related to the polyacrylamide percentage (concentration). For instance, a 7% polyacrylamide gel has larger pores than a 12% polyacrylamide gel. Low-percentage gels are used to resolve large proteins, and high-percentage gels are used to resolve small proteins. "Gradient gels" are specially prepared to have a low percentage of acrylamide at the top (beginning of sample path) and a high percentage at the bottom (end), enabling a broader range of protein sizes to be separated.
Electrophoresis gels are formulated in buffers that enable electrical current to flow through the matrix. The prepared solution is poured into the thin space between two glass or plastic plates that form a "cassette". This process is referred to as “casting a gel”. Once the gel polymerizes, the cassette is mounted (usually vertically) into an apparatus so that the top and bottom edges are placed in contact with buffer chambers containing a cathode and an anode, respectively. The running buffer contains ions that conduct current through the gel. When proteins are added in wells at the top edge and current is applied, the proteins are drawn by the current through the matrix slab and separated by the sieving properties of the gel.
To obtain optimal resolution of proteins, a “stacking” gel is cast over the top of the “resolving” gel. The stacking gel has a lower concentration of acrylamide (e.g., 7% for larger pore size), lower pH (e.g., 6.8), and a different ionic content. This allows the proteins in a loaded sample to be concentrated into one tight band during the first few minutes of electrophoresis before entering the resolving portion of a gel. A stacking gel is not necessary when using a gradient gel, as the gradient itself performs this function.
Polyacrylamide gel electrophoresis in progress. Prepared gel cassettes are added to a gel tank, in this case the Invitrogen Mini Gel Tank, which holds two mini gels at a time. After wells are loaded with protein samples, the gels submerged in a conducting running buffer, and electrical current is applied, typically for 20 to 40 minutes. Run times vary according to the size and percentage of the gel and particular gel chemistry.
Traditionally, researchers casted their own gels using standard recipes that are widely available in protein methods literature. Most laboratories now depend on the convenience and consistency afforded by commercially available, ready-to-use precast gels. Precast gels are available in a variety of percentages, including difficult-to-pour gradient gels that provide excellent resolution and that separate proteins over the widest possible range of molecular weights. Precast gels are also available with several different buffer formulations (e.g., Tris-glycine, Tris-acetate, Bis-Tris), which are designed to optimize shelf life, run time, and/or protein resolution.
For researchers who require unique gel formulations not available as precast gels, a wide range of reagents and equipment are available. However, technological innovations in buffers and gel polymerization methods enable manufacturers to produce gels with greater uniformity and longer shelf life than individual researchers can prepare on their own with traditional equipment and methods. In addition, precast polyacrylamide gels eliminate the need to work with the acrylamide monomer, which is a known neurotoxin and suspected carcinogen.
Precast vs. handcast protein gels for SDS-PAGE. Polyacrylamide gels can be purchased precast and ready to use (left) or prepared from reagents in the lab using a gel-casting system (right). Pictured here are the Novex WedgeWell 4-20% Tris-Glycine Mini Gels (left) and the SureCast Gel Handcast System. Although no universal standard exists among manufacturers, the term “mini gel” typically refers to gels that are 1 mm thick and approximately 8 x 8 cm. Cassette dimensions and compatibility between tanks varies. Most Invitrogen mini gel cassettes, including those shown here, are compatible with the Invitrogen Mini Gel Tank. Precast gels are also available in midi gel formats (not shown). The XCell4 SureLock Midi-Cell Electrophoresis System is designed to accommodate up to 4 midi-gels (8 x13 cm).
Example gel selection guide. Many types of Invitrogen electrophoresis gels are available to enable researchers to separate proteins by different properties or for specific applications. (Click the image to visit the Protein Gels page, where there is a product-linked version of this guide.)
SDS-PAGE & denaturing gels
SDS-PAGE is used for routine separation and analysis of proteins because of its speed, simplicity, and resolving capability. In SDS-PAGE, the gel is cast in a buffer containing sodium dodecyl sulfate (SDS), an anionic detergent. The protein samples are heated with SDS before electrophoresis so that the charge density of all proteins is made roughly equal. The SDS (aided by heat) denatures proteins in the sample and binds tightly to the uncoiled molecules. Usually, a reducing agent such as dithiothreitol (DTT) is also added to cleave protein disulfide bonds and ensure that no tertiary or quaternary protein structure remains. Consequently, when these samples are electrophoresed, proteins separate according to mass alone, with very little effect from compositional differences.
When a set of proteins of known mass are run alongside samples in the same gel, they provide a reference by which the mass of sample proteins can be determined. These sets of reference proteins are called mass markers or molecular weight markers (MW markers), protein ladders, or size standards, and they are available commercially in several forms.
In native PAGE, proteins are separated according to the net charge, size, and shape of their native structure. Electrophoretic migration occurs because most proteins carry a net negative charge in alkaline running buffers. The higher the negative charge density (more charges per molecule mass), the faster a protein will migrate. At the same time, the frictional force of the gel matrix creates a sieving effect, regulating the movement of proteins according to their size and three-dimensional shape. Small proteins face only a small frictional force, while larger proteins face a larger frictional force. Thus native PAGE separates proteins based upon both their charge and mass.
Because no denaturants are used in native PAGE, subunit interactions within a multimeric protein are generally retained and information can be gained about the quaternary structure. In addition, some proteins retain their enzymatic activity (function) following separation by native PAGE. Thus, this technique may be used for preparation of purified, active proteins.
Following electrophoresis, proteins can be recovered from a native gel by passive diffusion or electro-elution. To maintain the integrity of proteins during electrophoresis, it is important to keep the apparatus cool and minimize denaturation and proteolysis. pH extremes should generally be avoided in native PAGE, as they may lead to irreversible damage, such as denaturation or aggregation, to proteins of interest.
1-dimensional polyacrylamide gel electrophoresis
The most common form of protein gel electrophoresis is comparative analysis of multiple samples by one-dimensional (1D) electrophoresis. Gel sizes range from 2 x 3 cm (tiny) to 15 x 18 cm (large format). The most popular size (approx. 8 x 8 cm) is usually referred to as a "minigel". Medium-sized gels (8 x 13 cm) are called “midi gels”. Small gels require less time and reagents than their larger counterparts and are suited for rapid protein screening. However, larger gels provide better resolution and are needed for separating similar proteins or a large number of proteins.
Protein samples are added to sample wells at the top of the gel. When the electrical current is applied, the proteins move down through the gel matrix, creating what are called "lanes" of protein "bands". Samples that are loaded in adjacent wells and electrophoresed together are easily compared to each other after staining or other detection strategies. The intensity of staining and "thickness" of protein bands are indicative of their relative abundance. The positions (height) of bands within their respective lanes indicate their relative sizes (and/or other factors affecting their rate of migration through the gel).
Protein lanes and bands in 1D SDS-PAGE. Polyacrylamide gels may be formulated using a variety of different gel chemistries. Depicted here is a Protein ladders, purified proteins and E. coli lysate were loaded on a 4–20% gradient Novex WedgeWell Tris-Glycine gel; Lanes 1, 5, 10: 5 µL Thermo Scientific PageRuler Unstained Protein Ladder); lanes 2, 6, 9: 5 µL Mark12 Unstained Standard; lane 3: 10 µg E. coli lysate (10 µL sample volume); lane 4: 6 µg BSA (10 µL sample volume); lane 7: 6 µg hIgG (10 µL sample volume); lane 8: 20 µg E. coli lysate (20 µL sample volume). Electrophoresis was performed using the Mini Gel Tank. Sharp, straight bands were observed after staining with SimplyBlue SafeStain. Images were acquired using a flatbed scanner.
2-dimensional polyacrylamide gel electrophoresis
Multiple components of a single sample can be resolved most completely by two-dimensional electrophoresis (2D-PAGE). The first dimension separates proteins according to their native isoelectric point (pI) using a form of electrophoresis called isoelectric focusing (IEF). The second dimension separates by mass using ordinary SDS-PAGE. 2D PAGE provides the highest resolution for protein analysis and is an important technique in proteomic research, where resolution of thousands of proteins on a single gel is sometimes necessary.
To perform IEF, a pH gradient is established in a tube or strip gel using a specially formulated buffer system or ampholyte mixture. Ready-made IEF strip gels (called immobilized pH gradient strips or IPG strips) and required instruments are available from certain manufacturers. During IEF, proteins migrate within the strip to become focused at the pH points at which their net charges are zero. These are their respective isoelectric points.
The IEF strip is then laid sideways across the top of an ordinary 1D gel, allowing the proteins to be separated in the second dimension according to size.
Analysis of spinach chloroplast extract by two-dimensional electrophoresis (2DE).Spinach chloroplast extract was prefractionated in the ZOOM IEF Fractionator and the individual fractions were then separated by 2DE using narrow pH range ZOOM Strips and NuPAGE Novex 4–12% Bis-Tris ZOOM Gels. Gels were Coomassie stained using SimplyBlue SafeStain.
Protein samples prepared for SDS-PAGE analysis are denatured by heating in the presence of a sample buffer containing 1% SDS with or without a reducing agent such as 20mM DTT, 2-mercaptoethanol (BME) or TCEP. The protein sample is mixed with the sample buffer and heated or boiled for 3 to 5 minutes (according to the specific protocol) then cooled to room temperature before it is pipetted into the sample well of a gel. Loading buffers also contain glycerol so that they are heavier than water and sink neatly to the bottom of the buffer-submerged well when added to a gel.
If a suitable, negatively charged, low-molecular weight dye is also included in the sample buffer, it will migrate at the buffer-front, enabling one to monitor the progress of electrophoresis. The most common tracking dye for sample loading buffers is bromophenol blue. Thermo Scientific Lane Marker Sample Buffers contain a bright pink tracking dye.
Samples may contain substances that interfere with electrophoresis by adversely affecting the migration of protein bands in the gel. Substances such as guanidine hydrochloride and ionic detergents can result in protein bands that appear smeared or wavy. The Thermo Scientific Pierce SDS-PAGE Sample Prep Kit facilitates removal of these interfering components using a specialized affinity resin system. Methods such as this are much faster than dialysis, ultrafiltration, or acetone precipitation, and the protein recovery is generally higher.
Protein gel electrophoresis power requirements as well as separation and migration patterns are determined by the chemical composition and pH of the buffer system. Three basic types of buffers are required: the gel casting buffer, the sample buffer, and the running buffer that fills the electrode reservoirs. Electrophoresis may be performed using continuous or discontinuous buffer systems. A continuous buffer system, which utilizes only one buffer in the gel, sample, and gel chamber reservoirs, is most often used for nucleic acid analysis and rarely used for protein gel electrophoresis. Proteins separated using a continuous buffer system tend to be diffuse and poorly resolved. Conversely, discontinuous buffer systems utilize a different gel buffer and running buffer. These systems also use two gel layers of different pore sizes and different buffer compositions (the stacking and separating gels). Electrophoresis using a discontinuous buffer system results in concentration of the sample and higher resolution. The following table summarized various discontinuous gel buffer systems.
Common gel and buffer chemistries for discontinuous PAGE
|Gel Type||Sample||Running||Selection criteria|
|Tris-glycine, pH 8.6||Laemmli||Tris, glycine, and SDS (pH8.3)||Ease of preparation; relatively inexpensive|
|Bis-Tris, pH 6.4||Laemmli||Tris, MES or MOPs||Relatively long shelf life; room temperature storage; results cannot be compared to those obtained with Tris-glycine gels|
|Tris-Acetate, pH 7.0||Tris-Glycine SDS||Tris/Tricine/SDS||Superior separation of protein complexes and high MW proteins; relatively long shelf life|
|Tris-glycine , pH 8.6||Native||Tris/Glycine||Retention of native protein structure|
|Tris-acetate, pH 7.0||Tris-Glycine Native||Tris/Glycine||Superior separation of protein complexes and high MW proteins|
|Tris-Tricine||Tricine/SDS||Tris/Tricine/SDS||Ideal for separating peptides and low molecular weight proteins|
|IEF||IEF Sample Buffer||IEF cathode/IEF anode buffers||Use to separate proteins according to isoelectric point (pI) rather than molecular weight|
|Zymogram||Tris-Glycine SDS||Tris/glycine/SDS||Gelatin or casein gels provide substrates used to detect proteases|
Comparison of discontinuous gel systems
The classical Laemmli system, consisting of Tris-glycine gels and Tris-glycine running buffer, is employed for both SDS-PAGE and native PAGE and may be utilized to separate a broad range of proteins. This system is used widely because reagents for casting Tris-glycine gels are relatively inexpensive and readily available. Gels using this chemistry are also available in a variety of precast gel formats and percentages.
The formulation of this discontinuous buffer system creates a stacking effect to produce sharp protein bands at the beginning of the electrophoretic run. A boundary is formed between chloride, the leading ion, and glycinate, the trailing ion. Tris buffer provides the common cations. As proteins migrate into the resolving gel, they are separated according to size. Tris-glycine gels are used in conjunction with Laemmli sample buffer, and Tris/glycine/SDS running buffer is used for denaturing SDS-PAGE. Native PAGE is performed using native sample and running buffers without denaturants or SDS. The pH and ionic strength of the buffer used for running the gel (Tris, pH 8.3) are different from those of the buffers used in the stacking gel (Tris, pH 6.8) and the resolving gel (Tris, pH 8.8). The highly alkaline operating pH of the Laemmli system may cause band distortion, loss of resolution, or artifact bands.
- Hydrolysis of polyacrylamide at the high pH of the resolving gel, resulting in a short shelf life of 8 weeks
- Chemical alterations such as deamination and alkylation of proteins due to the high pH of the resolving gel
- Reoxidation of reduced disulfides from cysteine-containing proteins
- Cleavage of Asp-Pro bonds of proteins when heated at 100°C in Laemmli sample buffer, pH 5.2
In contrast to conventional Tris-glycine gels, Bis-Tris HCI–buffered (pH 6.4) gels are more acidic, thus offering enhanced stability and greatly extended shelf-life over Tris-glycine gels. For Bis-Tris gels, chloride serves as the leading ion and MES or MOPS act as the trailing ion. Bis-Tris buffer forms the common cation. Markedly different protein migration patterns are produced depending on whether a Bis-Tris gel is run with MES or MOPS denaturing running buffer: MES buffer is used for small proteins, and MOPS buffer is used for mid-sized proteins. Due to differences in ionic composition and pH, gel patterns obtained with Bis-Tris gels cannot be compared to those obtained with Tris-glycine gels. To prevent protein reoxidation, Bis-Tris gels must be run with alternative reducing agents such as sodium bisulfite. Reducing agents frequently used with Tris-glycine gels, such as beta-mercaptoethanol and dithiothreitol (DTT), do not undergo ionization at low pH levels and are not able to migrate with proteins in a Bis-Tris gel.
Tris-acetate gels are designed for separating large molecular weight proteins and may be used with both SDS-PAGE and native PAGE running buffers. Compared with Tris-glycine gels, Tris-acetate gels have a lower pH, which enhances the stability of these gels and minimizes protein modifications, resulting in sharper bands.
The Tris-Tricine gel system is a modification of the Tris-glycine gel system and is optimized to resolve low molecular weight proteins in the range of 2–20 kDa. As a result of reformulating the Laemmli running buffer and using Tricine in place of glycine, SDS-polypeptides form behind the leading ion front rather than running with the SDS front, thus allowing for their separation into discrete bands.
Isoelectric focusing (IEF) is a technique designed to separate proteins according to their isoelectric point (pI) rather than molecular weight. The pI is the pH at which a protein has no net charge and no longer moves in an electric field. IEF gels are run under native conditions and may be used for these applications: pI determination, detection of changes such as deamination, phosphorylation, or glycosylation, and resolution of different proteins similar in size that cannot be resolved using standard SDS-PAGE gels.
Zymogram gels are composed of gelatin or casein and are used to characterize proteases that utilize them as substrates. Samples are run under denaturing conditions, but due to the absence of reducing agents, proteins undergo renaturation. Proteolytic proteins present in the sample consume the substrate, generating clear bands against a background stained blue.
Protein gel electrophoresis chambers
To perform protein gel electrophoresis, the polyacrylamide gel and buffer must be placed in an electrophoresis chamber that is connected to a power source, and which is designed to conduct current through the buffer solution. When current is applied, the smaller molecules migrate more rapidly and the larger molecules migrate more slowly through the gel matrix. Multiple gel chamber designs exist. The choice of equipment is usually based on these factors: the dimensions of the gel cassette, with some tank designs accommodating more cassette sizes than others; the nature of the protein target, and corresponding gel resolution requirements; and whether a precast or handcast gel, and vertical or horizontal electrophoresis system, has been selected.
Mini gel tank for protein gel electrophoresis. This gel tank holds up to two mini gels and is compatible with the Invitrogen SureCast Gel Handcast System, and with NuPAGE, Bolt, or Novex mini gels. The unique tank design enables side-by-side gel loading and enhanced viewing during use.
Commercial gels are available in two size formats, for mini gels and midi gels. Both gels have similar run lengths, but midi gels are wider than mini gels, allowing midi gels to have more wells or larger wells. The additional wells in the midi gels permit more protein samples or large sample volumes to be loaded onto one gel.
Protein ladders and standards
To assess the molecular masses (sizes) of proteins in a gel, a prepared mixture containing several proteins of known molecular masses is run alongside the test sample in one or more lanes of the gel. Such sets of known proteins are called protein molecular weight (or mass) markers or protein ladders. A standard curve can be constructed from the distances migrated by each marker protein. The distance migrated by the unknown protein is then plotted, and the molecular weight is extrapolated from the standard curve.
Several kinds of ready-to-use protein molecular weight (MW) markers are available that are either unlabeled or prestained for different modes of detection. These are pre-reduced and, therefore, primarily suited for SDS-PAGE rather than native PAGE. MW markers can also be made detectable via specialized labels, such as a fluorescent tag, and by other methods.
SDS-PAGE band profile of the Thermo Scientific PageRuler Plus Prestained Protein Ladder. Images are from a 4–20% Tris-glycine gel (SDS-PAGE) and subsequent transfer to a membrane.
Once protein bands have been separated by polyacrylamide gel electrophoresis, they can be blotted (transferred) to a membrane for analysis by western blotting. Alternatively, they can be visualized directly in the gel using various staining or detection methods.
Coomassie dye is the most popular reagent for staining protein bands in electrophoretic gels. In acidic buffer conditions, Coomassie dye binds to basic and hydrophobic residues of proteins, changing in color from dull reddish-brown to intense blue. As with all staining methods, Coomassie dye reagents detect some proteins better than others based on their chemistry of action and differences in protein composition. For most proteins, however, Coomassie dye reagents detect as little as 10 ng per band in a mini gel.
In addition to the use of Coomassie dye, the traditional staining methods below may be employed for detection of protein bands in a gel:
|Gel Staining Method||Considerations|
Metallic silver deposited onto the surface of the gel is used to detect protein bands. Method can detect ≤0.5 nanograms of protein and chemically crosslinks proteins in the gel matrix. Formulations exist that may be suitable for downstream applications, such as mass spectrometry
Zinc reversible stains
Zinc does not stain the protein directly, but produces an opaque background with clear, unstained protein bands. Method detects ≤ 1 ng of protein and is reversible, allowing for analysis by mass spectrometry or western blotting
Fluorescent stains are available with excitation and emission maxima corresponding to the common filter sets and laser settings of most fluorescence imagers
Traditional staining methods involve one to several long incubation and wash steps. Recently, more rapid staining protocols have been developed using powered (electrophoretic) devices, such as the Thermo Scientific™ Pierce™ Power Stainer.
The Thermo Scientific Pierce Power Stainer. This powered device enables rapid (6–11 minute) Coomassie dye staining of proteins in polyacrylamide gels, including the removal of unbound stain, in a single step. The small, easy-to-use device consists of the Pierce Power Station and Pierce Power Stain Cassette, which accommodates up to two mini gels or one midi gel at a time. The staining procedure is designed exclusively for use with Pierce Power Staining Kits.
Transfer and western blots
Upon completion of protein gel electrophoresis, an immunodetection technique known as western blotting—also referred to as immunoblotting—may be performed. This technique involves the transfer of proteins that have been separated on a polyacrylamide gel to a solid support matrix, and subsequent detection of one or more specific proteins utilizing antigen-specific antibodies. Western blotting was introduced by Towbin et al. in 1979 and is now routinely used for protein analysis. Western blotting can produce qualitative and semiquantitative data about proteins of interest.
Following gel electrophoresis, proteins are transferred to a membrane in preparation for western blotting and other immunoblotting techniques.
Scientists have used a variety of methods to transfer size-separated proteins from a polyacrylamide gel to the membrane support, including diffusion transfer, capillary transfer, heat-accelerated convectional transfer, vacuum blotting, and electroblotting (electrotransfer). Among these methods, electroblotting has emerged as the most popular because it is faster and more efficient than the other methods. The technique employs the use of various types of membranes, buffers, stains, molecular weight markers, and transfer devices. When setting up a western blot experiment, one of the first decisions is the choice of electrotransfer method: wet, semi-dry, or dry. This choice will drive what transfer buffers (or absence thereof) will be used and may also impact the choice of membrane products, because some transfer devices have specific requirements. The choice of transfer device may also drive the gel size, blotting area, and throughput, as well as potentially impact transfer efficiencies.
- Coligan, J.E., et al., Eds. (2002). Electrophoresis, In Current Protocols in Protein Science, pp. 10.0.1-10.4.36. John Wiley and Sons, Inc. New York.
- Bollag, D.M., Rozycki, M.D. and Edelstein, S.J. (2002). Protein Methods, 2nd ed. Wiley-Liss, Inc. New York.
- Hames, B.D. and Rickwood, D. Eds. (1990) Gel Electrophoresis of Proteins: a Practical Approach, 2nd ed. Oxford University Press, New York.
- Laemmli, U.K. 1970. Nature. 227: 680-685. PMID: 5432063
For Research Use Only. Not for use in diagnostic procedures.