Overview of ELISA
What is ELISA (enzyme-linked immunosorbent assay)?
ELISA (enzyme-linked immunosorbent assay) is a plate-based assay technique designed for detecting and quantifying substances such as peptides, proteins, antibodies and hormones. Other names, such as enzyme immunoassay (EIA), are also used to describe the same technology. In an ELISA, an antigen must be immobilized on a solid surface and then complexed with an antibody that is linked to an enzyme. Detection is accomplished by assessing the conjugated enzyme activity via incubation with a substrate to produce a measureable product. The most crucial element of the detection strategy is a highly specific antibody-antigen interaction.
- ELISA formats (direct, sandwich, etc.)
- Direct vs. indirect detection ELISA strategies
- Other ELISA formats (competitive ELISPOT, etc.)
- Complete, ready-to-use ELISA kits
- Selecting and coating ELISA plates
- Pre-coated ELISA plates
- Antibodies and probes for ELISA
- Blocking buffers and wash buffers
- Detection strategies for ELISA
ELISAs are typically performed in 96-well (or 384-well) polystyrene plates, which passively bind antibodies and proteins. It is this binding and immobilization of reagents that makes ELISAs so easy to design and perform. Having the reactants of the ELISA immobilized to the microplate surface makes it easy to separate bound from non-bound material during the assay. This ability to wash away nonspecifically bound materials makes the ELISA a powerful tool for measuring specific analytes within a crude preparation.
A detection enzyme or other tag can be linked directly to the primary antibody or introduced through a secondary antibody that recognizes the primary antibody. It can also be linked to a protein such as streptavidin if the primary antibody is biotin labeled. The most commonly used enzyme labels are horseradish peroxidase (HRP) and alkaline phosphatase (AP). Other enzymes have been used as well, but they have not gained widespread acceptance because of limited substrate options. These include β-galactosidase, acetylcholinesterase and catalase. A large selection of substrates is available for performing ELISA with an HRP or AP conjugate. The choice of substrate depends upon the required assay sensitivity and the instrumentation available for signal-detection (spectrophotometer, fluorometer or luminometer).
ELISAs can be performed with a number of modifications to the basic procedure. The key step, immobilization of the antigen of interest, can be accomplished by direct adsorption to the assay plate or indirectly via a capture antibody that has been attached to the plate. The antigen is then detected either directly (labeled primary antibody) or indirectly (labeled secondary antibody). The most powerful ELISA assay format is the sandwich assay. This type of capture assay is called a “sandwich” assay because the analyte to be measured is bound between two primary antibodies – the capture antibody and the detection antibody. The sandwich format is used because it is sensitive and robust.
Diagram of common ELISA formats (direct vs. sandwich assays). In the assay, the antigen of interest is immobilized by direct adsorption to the assay plate or by first attaching a capture antibody to the plate surface. Detection of the antigen can then be performed using an enzyme-conjugated primary antibody (direct detection) or a matched set of unlabeled primary and conjugated secondary antibodies (indirect detection).
Direct vs. indirect detection ELISA strategies
Among the standard assay formats discussed and illustrated above, where differences in both capture and detection were the concern, it is important to differentiate between the particular strategies that exist specifically for the detection step. Irrespective of the method by which an antigen is captured on the plate (by direct adsorption to the surface or through a pre-coated "capture" antibody, as in a sandwich ELISA), it is the detection step (as either direct or indirect detection) that largely determines the sensitivity of an ELISA.
Watch this video about ELISA detection and signal-amplification strategies
The direct detection method uses a labeled primary antibody that reacts directly with the antigen. Direct detection can be performed with an antigen that is directly immobilized on the assay plate or with the capture assay format. Direct detection while not widely used in ELISA is quite common for immunohistochemical staining of tissues and cells.
The indirect detection method uses a labeled secondary antibody for detection and is the most popular format for ELISA. The secondary antibody has specificity for the primary antibody. In a sandwich ELISA, it is critical that the secondary antibody be specific for the detection primary antibody only (and not the capture antibody) or the assay will not be specific for the antigen. Generally, this is achieved by using capture and primary antibodies from different host species (e.g., mouse IgG and rabbit IgG, respectively). For sandwich assays, it is beneficial to use secondary antibodies that have been cross-adsorbed to remove any secondary antibodies that might have affinity for the capture antibody.
|Direct ELISA detection|
|Indirect ELISA detection|
Fluorescent tags and other alternatives to enzyme-based detection can be used for plate-based assays. Despite not involving reporter-enzymes, these methods are also generally referred to as a type of ELISA. Likewise, wherever detectable probes and specific protein binding interactions can be used in a plate-based method, these assays are often called ELISAs despite not involving antibodies.
Other ELISA formats
Besides the standard direct and sandwich formats described above, several other styles of ELISA’s exist:
Competitive ELISA is a strategy that is commonly used when the antigen is small and has only one epitope, or antibody binding site. One variation of this method consists of labeling purified antigen instead of the antibody. Unlabeled antigen from samples and the labeled antigen compete for binding to the capture antibody. A decrease in signal from the purified antigen indicates the presence of the antigen in samples when compared to assay wells with labeled antigen alone.
Watch this video about competitive ELISA methods
ELISPOT (enzyme-linked immunospot assay) refers to ELISA-like capture and measurement of proteins secreted by cells that are plated in PVDF-membrane-backed microplate wells. It is a "sandwich" assay in which the proteins are captured locally as they are secreted by the plated cells, and detection is with a precipitating substrate. ELISPOT is like a western blot in that the result is spots on a membrane surface.
In-cell ELISA is performed with cells that are plated and cultured overnight in standard microplates. After the cultured cells are fixed, permeabilized and blocked, target proteins are detected with antibodies. This is an indirect assay, not a sandwich assay. The secondary antibodies are either fluorescent (for direct measurement by a fluorescent plate reader or microscope) or enzyme-conjugated (for detection with a soluble substrate using a plate reader).
ELISA is nearly always performed using 96-well or 384-well polystyrene plates and samples in solution (i.e., biological fluids, culture media or cell lysates). This is the platform discussed in the remainder of this article.
- ELISA Development and Optimization
- Spike-and-Recovery and Linearity-of-Dilution Assessment for ELISA
- Factors Affecting Signal Generation in ELISA
- ELISA Protocols
- ELISA Troubleshooting Guide
- Antibodies and Immunoassays Support Center
- ELISA Kits and Antibody Pairs Support Center
- Invitrogen Antibody Validation
- Antibodies Learning Center
This 64-page guide provided detailed information about different tools for protein quantitation. Download this valuable technical resource that covers technologies useful for cancer and inflammation research, immunology, neurology and more. Learn more about how antibody pairs, ELISA kits, and multiplex kits for the Invitrogen Luminex platform may help advance your research.
Complete, ready-to-use ELISA kits
In addition to the individual components and general principles of ELISA discussed in the remainder of this article, ready-to-use sandwich ELISA kits are commercially available for detection of hundreds of specific cytokines, neurobiology analytes and phosphorylated proteins that are common targets of research interest.
For many targets, two kit types are available:
- ELISA kits contain pre-coated antibody-plates, detection antibodies, buffers, diluents, standards, and substrates. In addition to traditional ELISA kits with pre-coated plates include only the capture antibody when the sample is added, Thermo Fisher Scientific offers Instant ELISA kit plates that contain all of the necessary components including capture antibody and lyophilized detection antibody, streptavidin-HRP, and sample diluent. In addition, strip wells containing the standard for the standard curve are provided separately.
Comparison of instant ELISA technology vs. conventional ELISA procedures. In contrast to conventional ELISA kits, Thermo Scientific Invitrogen Instant ELISA kits were produced to include both the capture antibody and lyophilized detection antibody and other reagents required to develop an ELISA.
- Antibody pair kits contain only matched antibodies and standard (no plates or detection reagents).
This ELISA format selection guide compares characteristics of Thermo Fisher Invitrogen antibody pair kits and ELISA kits.
|Build-it-yourself||Uncoated ELISA kits||Coated ELISA kits*||Instant ELISA kits|
|Ready-to-use reagents||No, an overnight coating process is required||Yes, an overnight coating process is required||Yes||Yes|
|Analytical sensitivity**||<10 pg/ml||<10 pg/ml||<10 pg/ml||<5 pg/ml|
|Dynamic range**||<5-250 pg/ml||<5-250 pg/ml||<5-250 pg/ml||<7.8-500 pg/ml|
|Incubation time**||4 h||3.5-4 h||2,5-4 h||2-3 h|
|Number of targets measured per well||1||1||1||1|
|Readout||HRP-TMB (colorimetric)||HRP-TMB (colorimetric)||HRP-TMB (colorimetric)||HRP-TMB (colorimetric)|
|Instrumentation needed||Microplate reader, absorbance||Microplate reader, absorbance||Microplate reader, absorbance||Microplate reader, absorbance|
|2 min||2 min||2 min||2 min|
*Values in this table refer to our Standard Colorimetric kits. Ultrasensitive kits are also available.
**Every assay has its oven specifications. Please consult the protocol for your specific immunoassays/kits.
ELISA format selection guide. Characteristics of Thermo Fisher Invitrogen antibody pair kits and ELISA kits.
Selecting and coating ELISA plates
When developing a new ELISA for a specific antigen, the first step is to optimize the plate-coating conditions for the antigen or capture antibody. Begin by choosing an assay microplate (not tissue culture treated plates) with a minimum protein-binding capacity of 400 ng/cm². It is also important that the CV value (coefficient of variation) of the protein binding be low (<5% is preferred) so that there is limited deviation in values that should be identical in the assay results between wells and plates. The choice of plate color depends upon the signal being detected. Clear polystyrene flat bottom plates are used for colorimetric signals while black or white opaque plates are used for fluorescent and chemiluminescent signals. Visually inspect plates before use as imperfections or scratches in the plastic will cause aberrations when acquiring data from the developed assay.
Uncoated ELISA plates. Thermo Scientific Uncoated ELISA Plates are available with a variety of surfaces to optimize coating with the macromolecule of your choice. These plates are designed to deliver optimal results, lot-to-lot reliability and well-to-well reproducibility.
Plate coating is achieved through passive adsorption of the protein to the plastic of the assay microplate. This process occurs though hydrophobic interactions between the plastic and non-polar protein residues. Although individual proteins may require specific conditions or pretreatment for optimal binding, the most common method for coating plates involves adding a 2-10 μg/ml solution of protein dissolved in an alkaline buffer such as phosphate-buffered saline (pH 7.4) or carbonate-bicarbonate buffer (pH 9.4). The plate is left to incubate for several hours to overnight at 4-37° C. Typically, after removing the coating solution, blocking buffer is added to ensure that all remaining available binding surfaces of the plastic well are covered (see subsequent discussion). Coated plates can be used immediately or dried and stored at 4° C for later use, depending on the stability of the coated protein.
It is important to note that optimal coating conditions and plate binding capacity can vary with each protein and must be determined experimentally. With the exception of competition ELISAs, the plates are coated with more capture protein than can actually be bound during the assay in order to facilitate the largest working range of detection possible. Some proteins, especially antibodies, are best coated on plates at a concentration lower than the maximum binding capacity in order to prevent nonspecific binding in later steps by a phenomenon called "hooking". Hooking results from proteins getting trapped between the coating proteins which prevents effective washing and removal of unbound proteins. When hooking nonspecifically traps detection primary and secondary antibodies, high background signal results, lowering the signal to noise ratio and thus sensitivity of an assay. The following example illustrates how variations in polymer coatings may impact protein binding capacities.
IgG Binding on modified surfaces. The introduction of functional groups will affect the binding characteristics of the plastic polymer. This experiment demonstrates that surface modifications will affect binding of proteins. Comparison of adsorption of various proteins on non-treated control, Thermo Scientific Nunc MultiSorp Thermo Scientific Nunc MultiSorp and MaxiSorp flat-bottom plates indicates the importance of surface selection on assay optimization. Various molecules behave in distinctly different manners depending on the characteristics of the surface. For example, under basic conditions, IgG will adsorb to MaxiSorp modified polystyrene with significantly more capacity when compared with a non-treated control plate. In the case of MultiSorp, the functional groups on the surface restrict the protein absorption of IgG; evident by a binding capacity compared to the non-treated plate.
Pre-coated ELISA plates
For antibodies and proteins, coating plates by passive adsorption usually works well. However, problems can arise from passive adsorption, including improper orientation, denaturation, poor immobilization efficiency and binding of contaminants along with the target molecule. Antibodies can be attached to a microplate through the Fc region using Protein A, G, or A/G coated plates, which orients them properly and preserves their antigen binding capability. Fusion proteins can be attached to a microplate in the proper orientation using glutathione, metal-chelate, or capture-antibody coated plates. Peptides and other small molecules, which typically do not bind effectively by passive adsorption, can be biotinylated and attached with high efficiency to a streptavidin or NeutrAvidin protein coated plate. Biotinylated antibodies also can be immobilized on plates precoated with biotin-binding proteins. Using pre-coated plates in this manner physically separates the antigen or capture antibody from the surface of the plate as a protection from its denaturing effects.
There is a wide selection of high-performance surface coated plates (pre-coated and pre-blocked) in 96-well and 384-well format (black, clear or white). These coated microplates can be used for ELISA development and other plate-based assays with standard or fluorescent plate readers.
Antibodies and probes for ELISA
Either monoclonal or polyclonal antibodies can be used as the capture and detection antibodies in sandwich ELISA systems. Monoclonals have an inherent monospecificity toward a single epitope that allows fine detection and quantitation of small differences in antigen. A polyclonal is often used as the capture antibody to pull down as much of the antigen as possible. Then a monoclonal is used as the detecting antibody in the sandwich assay to provide improved specificity. In addition to the use of traditional monoclonal antibodies, recombinant monoclonal antibodies may also be utilized for ELISA. For example, Invitrogen rabbit recombinant antibodies are derived from antibody-producing cell lines engineered to express specific antibody heavy and light chain DNA sequences. Compared to traditional monoclonal antibodies derived from hybridomas, recombinant antibodies are not susceptible to cell-line drift or lot-to-lot variation, thus allowing for peak antigen specificity. The example that follows presents data produced using a recombinant rabbit monoclonal antibody.
Sandwich ELISA. This experiment was performed using ABfinity EGF Recombinant Rabbit Monoclonal Antibody at 2 µg/mL. A standard curve was plotted with full length recombinant EGF protein with concentrations ranging from 0.3 pg/mL to 12.5 ng/mL. An anti-EGF antibody conjugated to biotin was used as a detector at a concentration of 2 µg/mL.
An important consideration in designing a sandwich ELISA is that the capture and detection antibodies must recognize two different non-overlapping epitopes. When the antigen binds to the capture antibody, the epitope recognized by the detection antibody must not be obscured or altered. Capture and detection antibodies that do not interfere with one another and can bind simultaneously are called "matched pairs" and are suitable for developing a sandwich ELISA. Many primary antibody suppliers provide information about epitopes and indicate pairs of antibodies that have been validated in ELISA as matched pairs.
Another design consideration in choosing antibodies is cost. A polyclonal antibody is generally less expensive (~5 fold) to produce than a monoclonal. The specificity gained by using monoclonals for both the capture and detecting antibody must be weighed against the cost and time required for producing two monoclonal antibodies. Preparing a “self-sandwich” ELISA assay, where the same antibody is used for the capture and detection, can limit the dynamic range and sensitivity of the final ELISA.
Blocking buffers and wash buffers
The binding capacity of microplate wells is typically higher than the amount of protein coated in each well. The remaining surface area must be blocked to prevent antibodies or other proteins from adsorbing to the plate during subsequent steps. A blocking buffer is a solution of irrelevant protein, mixture of proteins, or other compound that passively adsorbs to all remaining binding surfaces of the plate. The blocking buffer is effective if it improves the sensitivity of an assay by reducing background signal and improving the signal-to-noise ratio. The ideal blocking buffer will bind to all potential sites of nonspecific interaction, eliminating background altogether, without altering or obscuring the epitope for antibody binding.
When developing any new ELISA, it is important to test several different blockers for the highest signal: noise ratio in the assay. Many factors can influence nonspecific binding, including various protein: protein interactions unique to the samples and antibodies involved. The most important parameter when selecting a blocker is the signal: noise ratio, which is measured as the signal obtained with a sample containing the target analyte as compared to that obtained with a sample without the target analyte. Using inadequate amounts of blocker will result in excessive background and a reduced signal: noise ratio. Using excessive concentrations of blocker may mask antibody-antigen interactions or inhibit the enzyme, again causing a reduction of the signal: noise ratio. No single blocking agent is ideal for every occasion and empirical testing is essential for true optimization of the blocking step.
In addition to blocking, it is essential to perform thorough washes between each step of the ELISA. Washing steps are necessary to remove nonbound reagents and decrease background, thereby increasing the signal: noise ratio. Insufficient washing will allow high background, while excessive washing might result in decreased sensitivity caused by elution of the antibody and/or antigen from the well. Washing is performed in a physiologic buffer such as Tris-buffered saline (TBS) or phosphate-buffered saline (PBS) without any additives. Usually, a detergent such as 0.05% Tween-20 is added to the buffer to help remove nonspecifically bound material. Another common technique is to use a dilute solution of the blocking buffer along with some added detergent. Including the blocking agent and adding a detergent in wash buffers helps to minimize background in the assay. For best results, use high-purity detergents to prevent introduction of impurities that will interfere with the assay such enzyme inhibitors or peroxides.
Detection strategies for ELISA
The final stage in all ELISA systems is a detection step. Unless a radioactive or fluorescent tag was used, this involves the introduction of an enzyme substrate. The enzyme converts the substrate to a detectable product. If an ELISA has been constructed and developed properly, then the intensity of signal produced when the substrate is added will be directly proportional to the amount of antigen captured in the plate and bound by the detection reagents. Enzyme-conjugated antibodies (especially those involving horseradish peroxidase, HRP) offer the most flexibility in detection and documentation methods for ELISA because of the variety of substrates available for chromogenic, chemifluorescent and chemiluminescent imaging. The following illustration describes the chemical reaction associated with luminol, a reagent that exhibits chemiluminescence.
Luminol reaction. This illustrates the chemiluminescent reaction used by the Thermo Scientific SuperSignal ELISA Pico Substrate for HRP.
Though not as sensitive as fluorescent or chemiluminescent substrates, chromogenic ELISA substrates allow direct visualization and enable kinetic studies to be performed. Furthermore, chromogenic ELISA substrates are detected with standard absorbance plate readers common to many laboratories. Fluorescent ELISA substrates are not as common and require a fluorometer that produces the correct excitation beam to cause signal emission to be generated from the fluorescent tag. Though best used with a luminometer plate reader, chemiluminescent substrates can be detected by various means including digital camera systems. Once drawback of using chemiluminescent substrates for ELISA is the signal intensity can vary more than with other substrates. For assays requiring many plates to be read, this can present a problem if the signal begins to decay before plates are read. For this reason, it is important to make sure the assay has been optimized with the substrate in order to avoid misinterpreting signal-fade in a sample as low antigen abundance. In the representative experiment that follows, the performance of multiple TMB (3, 3’, 5, 5’-tetramethylbenzidine)-based kits were compared. TMB, a common chromogenic substrate for HRP, yields a blue color when oxidized. In the example that follows, the performance of multiple TMB substrates are compared using ELSIA.
Comparison of sensitivities of various TMB ELISA Substrates Thermo Scientific 1-Step Ultra TMB-ELISA Substrate Solution.
- John R. Crowther, Methods in Molecular Biology, The ELISA Guidebook. Second Edition. Humana Press, a part of Springer Science + Business Media, LLC 2009.
- Butler J.E. The Behavior of Antigens and Antibodies Immobilized on a Solid Phase. In: M.H.V. Van Regenmortel, ed. Structure of Antigens. Boca Raton, FL: CRC Press, 1992: 209-259. Vol.1, 209; CRC Press, Inc.
- Lequin, Rudolf M. "Enzyme immunoassay (EIA)/enzyme-linked immunosorbent assay (ELISA)." Clinical chemistry 51.12 (2005): 2415-2418. (Short review paper)
- (Above references this seminal publication: Engvall, Eva, and Peter Perlmann. "Enzyme-linked immunosorbent assay (ELISA) quantitative assay of immunoglobulin G." Immunochemistry 8.9 (1971): 871-874.)
For Research Use Only. Not for use in diagnostic procedures.